> Error Code
> Gpib Configuration Utility
Gpib Configuration Utility
The system returned: (22) Invalid argument The remote host or network may be down. This error occurs when you try to take advantage of a driver feature that is either not implemented in the driver or is not currently usable. The error also can be due to a hardware limitation, such as when the DMA controller cannot address your buffer and the driver cannot remap the buffer. 9 DMA hardware uP This error occurs while sending out GPIB command bytes.
Generated Thu, 24 Nov 2016 06:15:05 GMT by s_wx1196 (squid/3.5.20) ERROR The requested URL could not be retrieved The following error was encountered while trying to retrieve the URL: http://0.0.0.6/ Connection What Our Users Say Press & Media Contacts Advertising DMCA Policy Brands × Login Login to ManualsLib Don't have an account? YourFeedback! My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation.
Gpib Configuration Utility
Restore password × Upload manual upload from disk upload from url Thank you for your help! For example, the "*IDN?" message from the previous example is only understood by IEEE 488.2 compliant instruments. This error occurs when a data-byte write operation, such as ibwrt, is attempted with no addressed listeners on the bus or if a command-byte operation, such as ibcmd or SendCmds, is
If you receive all of the data and get an EABO error, then look for a particular end of string character (e.g., linefeed or carriage return) and configure the GPIB board The message to the instrument may contain a command that the instrument does not understand. ECIC (1)Error Condition: Function requires the GPIB board to be the Controller-In-Charge (CIC).Description: Certain functions require your GPIB board to be the CIC - these functions are noted in the NI-488.2 Gpib Tutorial It is also returned when a process tries to unlock an interface when no such lock exists.Solutions: The way to avoid the ELCK error is to wait for a random amount
Often instruments will only respond to the standard commands after they have been placed in this mode, and otherwise they will give an error despite being sent a valid command. How To Find Gpib Address The instrument may use a particular EOS (end of string) character as its termination method, but you may forget to append this termination character to your message. The error can be due to an operating system error, such as when the operating system is unable to provide a physical address to the driver. have a peek here There are 14 possible error codes.
If it does not appear, then you can call the ibwait function (set the wait mask for the CIC bit) to delay further processing until control is passed to your GPIB Gpib Commands If your GPIB board is the System Controller, then use the ibsic function (or SendIFC function) to send an interface clear. This error occurs when there is a problem with a table used by a driver function. Add Comments 1 2 3 4 5 My Profile|Privacy|Legal|Contact NI© National Instruments Corporation.
How To Find Gpib Address
Poor|Excellent Yes No Document Quality? Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase English 100 ratings: 3.27 out of 5   GPIB Error Codes and Common Solutions Gpib Configuration Utility Answered Your Question? 1 2 3 4 5 Document needs work? Gpib Interactive Control Call ibpad 0 and ibsad 0 at the beginning of your program to properly configure your board’s address.
If EOS is the termination method, then be sure to append the termination character to the end of your message. This error indicates that this callback notification failed to rearm itself by returning an illegal value or when a fatal driver error (EDVR) has occurred.Solutions: Ensure that the value being returned The system returned: (22) Invalid argument The remote host or network may be down. This error can occur under several circumstances. Labview Error Code
For example, "*IDN?" is a common identification query for IEEE 488.2 compliant instruments. Make sure that the GPIB board is addressed correctly before calling ibrd, ibwrt, RcvRespMsg, or SendDataBytes. In this case, the shadow handshake is not possible and the error is returned to notify you of that fact. For example, if you disabled hardware DMA by removing the DRQ/DACK jumpers on her legacy AT-GPIB/TNT, a call to ibconfigIbcDMA with a value of 1 to enable DMA would return this
You should (almost) always leave your GPIB board configured for primary address 0 and no secondary address. Gpib-usb-hs Driver FindRQS and AllSpoll return ETAB if the list of addresses to serial poll is empty or none of the devices in the list are asserting SRQ. 30 No GPIB address input. Some scenarios are listed below: A valid board handle is passed in as a device handle parameter or vice versa An invalid board or device descriptor is passed as input to
This error occurs when a driver function is called with an invalid parameter. 5 Command requires GPIB Controller to be System Controller.
EADR is also returned by the function ibgts when the shadow handshake feature is requested and the GPIB ATN line is already unasserted. Generated Thu, 24 Nov 2016 06:15:05 GMT by s_wx1196 (squid/3.5.20) ERROR The requested URL could not be retrieved The following error was encountered while trying to retrieve the URL: http://0.0.0.8/ Connection Please Contact NI for all product and support inquiries. What Is Gpib Please try the request again.
My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. EDVR (0) ECIC (1) ENOL (2) EADR (3) EARG (4) ESAC (5) EABO (6) ENEB (7) EDMA (8) EOIP (10) ECAP (11) EFSO (12) EBUS (14) ESTB (15) ESRQ (16) ETAB EARG (4)Error Condition: Invalid argument to a function call.Description: EARG results when an invalid argument is passed to a function call. Please try the request again.
Your cache administrator is webmaster. Refer to the KnowledgeBase for more information about correcting errors in LabVIEW. Also make sure that processes do not lock an interface up for the entire period of execution.EARM (22)Error Condition: ibnotify callback failed to rearm.Possible Cause: This error occurs when we use Suggested Decimal Mnemonic EDVR ECIC ENOL EADR EARG ESAC EABO ENEB EOIP completed ECAP EFSO EBUS ESTB ESRQ A description of each error and some conditions under which it may occur
Avoid calling ibgts, except immediately after an ibcmd call. Code Description 0 Error connecting to driver or device. Device-level functions return ENEB when the specified access board of a device cannot be found in the system, even if the access board is configured in ibconf. 8 DMA hardware error Board-level functions return ENEB when the specified interface is configured in ibconf but cannot be found in the system.
The system returned: (22) Invalid argument The remote host or network may be down. This error occurs when the driver is unable to wait for the RQS bit to set during a device-level ibwait. 17 Unrecognized command. 19 Board not present. 20 Table error. For example, ibrd returns EADR if the interface is not addressed as a listener. The error code is returned in the variable IBERR.
Section Four × Show quick links Table of Contents Ibclr(bd) ClearingTheGpib ClearingTheDevice Ibtrg(bd) Ibloc(bd) PlacingADeviceInRemoteMode Previous page Next page 1...636465666768697071 Related Manuals for National Instruments GPIB-PC Desktop National Instruments MXI Verify that you are using the correct termination method for your instrument. Possible Cause: Often EDVR errors are caused when you misspell the interface name of your board or the device name of your instrument. The ibpad function will return what the previous setting for the device was, and you can check to see if the configured address matches the device's actual address.